- Open Access
- Open Peer Review
Clinical comparison of liquid-based and conventional cytology of oral brush biopsies: a randomized controlled trial
© The Author(s). 2018
- Received: 17 January 2018
- Accepted: 15 May 2018
- Published: 29 May 2018
Exfoliative cytology performed on oral brush samples can help dentists to decide, whether a given oral lesion is (pre-) malignant. The use of non-invasive brush biopsies as an auxiliary tool in the diagnosis of oral mucosal lesions has gained renewed interest since improvements in cytological techniques such as the development of adjuvant diagnostic tools and liquid-based cell preparation techniques.
The aim of this study was to compare the quality of two different preparation techniques (cell collectors): the conventional transfer procedure to glass slides and the so-called liquid-based cytology preparation method. Cell smears were collected from 10 orally healthy individuals (mean age: 24 years) from the palatine mucosa at two different times (baseline and 4 weeks later). Slides of both techniques were stained by Giemsa (n = 40) and May-Gruenwald Giemsa (n = 40). The statistical analysis was performed with Excel.
On specimen analysis, the liquid-based cytology showed statistically significant improvement compared to conventional glass sides (p < 0.001). Thin layers, which were performed by liquid-based cytology showed significantly better results in the parameters (p < 0.001): uniform distribution, cellular overlapping, cellular disformation, mucus, microbial colonies and debris. The conventional glass slides approach showed more cell overlapping and contamination with extraneous material than thin layers, which were performed by Orcellex® Brush cell collectors.
Both techniques are diagnostically reliable. The liquid-based method showed an overall improvement on sample preservation, specimen adequacy, visualization of cell morphology and reproducibility.
Liquid-based cytology simplifies cell collection due to easier handling and less transfer errors by dentists.
- Collection methods
- Preparation methods
- Oral brush biopsy
- Liquid-based cytology
- Conventional cytology
Oral brush biopsies were routinely used for cytological examination of the oral mucosa in the last decade [1–4]. The main advantages of oral brush cytology are: simple handling, well-tolerated, minimally invasive, and relatively painless for patient [5, 6]. The use of non-invasive brush biopsies as an auxiliary tool in the diagnosis of oral mucosal lesions has gained renewed interest since improvements in cytological techniques such as the development of adjuvant diagnostic tools and liquid-based cell preparation techniques [7–12]. Liquid-based cytology was originally developed to provide a near-monolayer of superficial cervicals cells by means of specimen filtration . Results from uterine cervix examinations have shown that the liquid-based preparations reduce problems related to sampling error, poor transfer of the harvested cells by clinicians and fixation of cellular samples [7, 14–17]. The liquid-based cytology allows immediate fixation of cells while removing unwanted harvested material e.g. blood cells, mucus and debris. This preparatory technique produces so-called thin-layers, showing a clear background with evenly dispersed cells and producing more homogenous samples than conventional smears. Liquid-based techniques also reduce the proportion of unsatisfactory samples, thereby diminishing false negative results [7, 14–16]. The technique employs mucosal cell collection by plastic devices, which are then placed in a vial containing preservative liquid before being transported to the laboratory where collected liquid-based cytology samples are processed to produce slide preparations . This is unlike conventional preparation in which exfoliated cells are spread onto a glass slide for cytological evaluation.
Sampling was carried out using cell collectors specially developed for the oral cavity [1, 18]. The requirements requested on cell collectors for oral smears are an optimal size and shape to provide access to the various areas of the oral cavity . Cells of the superficial layers as well as middle and deep cell groups of the epithelium can be obtained with the collector. The preparations usually contain superficial as well as intermediate cells, basal cells occur only in isolated cases [1, 19–21]. The literature represent a number of brushes have been utilized for collecting oral mucosal cell. The review from Alsarraf et al. shows that the most commonly used brush is the cytobrush followed by the OralCDx® brush . The most recent of all assessed brushes is the new custom-designed Orcellex® brush for use in the oral cavity .
The combination of a special plastic brush (Orcellex®) using liquid-based cytology shows current a potential method for early detection of oral cancer and precancer .
In the literature heterogeneous sampling techniques and brush cytology tools have been utilized for oral mucosal sampling. There are only a few comparative studies, which compare the conventional cytology with the liquid-based cytology for oral mucosa samples [7, 24].
Exfoliative cytology performed on oral brush samples can help dentists and other physicians to decide, whether a given oral lesion is (pre-) malignant or at an early and thus curable stage [2, 25, 26].
The aim of this prospective randomized study was to compare liquid-based and conventional cytology of oral brush biopsies in a standardized clinical experimental setting.
This study was endorsed by the Leipzig Ethics Committee (file: 030–14-27,012,014). The study corresponds to the Consort guidelines . In this study, the Helsinki Declaration was considered . Ten test persons were included in the prospective randomized clinical study (female n = 3, male n = 7, mean age 24 years). The inclusion criteria were: age from 20 to 30 years, healthy, no mucosa lesions.
The conventional collection method, PapCone® Brushes (Otto Bock PUR LifeScience GmbH, Duderstadt, Germany) were used to obtain a cell smear from both quadrants. In order to perform the brush biopsies, the test person’s tongue was held by the assistant with a dental mirror, and the cell collecting was performed by the investigator on both sides at the corresponding mucosal areas. During the obtainment, it was necessary to ensure uniform contact between all bristles of the cell collector and the oral mucosa at the notch in the splint [29, 30]. The cell collector was rotated ten times about its own axis under moderate pressure on the mucosa [18, 31]. Subsequently the collected cells were transferred directly onto glass slides (Thermo Scientific™ SuperFrost©, Gerhard Menzel B.V. & Co. KG, Braunschweig, Germany) by rotating the sponge on it. For a homogeneous cell smear, in which the cells lie in one layer, the uniform rolling motion is decisive [32–34]. The slides were air-dried. In the laboratory the fixation was carried out by placing the glass slides in alcohol containing glass cuvettes for 20 min (methanol 99.8 vol.%, Th. Geyer GmbH & Co. KG, Renningen, Germany). After drying, Giemsa’s azur-eosin-methylene blue solution (Merck KGaA, Darmstadt, Germany) was used for staining the glass slides in according to the manufacturer guidelines. After further air-drying the glass slides were covered with Entellan (Merck, Darmstadt, Germany) and coverslips. They were stored at room temperature until evaluation.
For liquid-based cytology the cell collection method was carried out using Orcellex® Brushes (Rovers Medical Devices B.V., Oss, The Netherlands) according to the procedure described above.
One brush per sampling site was used to obtain a representative amount of epithelial cells. Following the brush biopsies by Orcellex® Brushes, fixation was carried out by separating the brush head from the applicator and placing it into the fixation liquid in the BD SurePath™ Collection Vial (BD Diagnostics - TriPath, Erembodegem, Belgium) . In the laboratory of the Institute of Pathology (Mathias-Hospital, Rheine, Germany) the preparation of thin-layers (or monolayers) was done by the preparation procedure for SurePath™ preparations, which was originally developed for cervical smears . Thin layers with a so-called settling chamber with a diameter of 13 mm were produced according to the manufacturer guidelines [19, 35]. The automatic staining of the preparations was carried out using the staining solutions (May-Gruenwald’s eosin-methylene blue solution and Giemsa’s azur-eosin-methylene blue solution, both Merck KGaA, Darmstadt, Germany) according to the manufacturer’s guidelines. Afterwards, the glass slides were covered and stored at room temperature until evaluation. All slides were examined by the same operator . For comparative analysis of both techniques seven parameters were used: distribution, cellular overlapping, cellular disformation, mucus, blood cells, microbial colonies, debris. The categories were analyzed with “yes” (1) or “no” (0), and sum scores were calculated.
The individual scores for each of the seven parameters and the sum scores were tested for equality of variance using the F-Test. In the next step, the T-test was used to determine statistical differences between the two techniques. Significance was set at p < 0.05.
This study focuses on the cytological results of the two cell collectors and cell preparation methods. The probands of this study were oral healthy. No pathological cell deformations were observed.
Comparison of different staining methods
Giemsa’s azur-eosin-methylene blue solution
May-Gruenwald’s eosin-methylene blue solution modified, Giemsa’s azur-eosin-methylene blue solution
staining cell nucleus
reddish - violet
from bluish to reddish - violet
staining microbial colonies
Pink to reddish
reddish to purple
blue to purple
blue to pink
The smears were performed by two different cell collectors. The glass slides made with the PapCone® Brushes differ significantly from the smears taken with the Orcellex® Brush.
Sum score and significance
liquid based cytology
p < 0.001
p < 0.001
p < 0.001
p < 0.001
p = 0.160
p < 0.001
p < 0.001
p < 0.001
The preparations which were produced with the liquid-based cell preparation method showed a uniform distribution of the cells in the circular evaluation area, even if there were glass slides with more and with fewer cells present. In addition, superpositions of the cells with other cells or with themselves occurred very rarely. There was hardly any foreign material compared to the smears with the PapCone® Brushes.
Results from this study have shown that the liquid-based cytology can offer significant advantages over the conventional exfoliative cytology in clinical practice.
First, some aspects of the materials and methods used will be discussed. Cell smears were taken from the selected areas of the palatine mucosa at two different times (first smear and four weeks later). In order to be independent of the current state of the cell cycle this time interval was chosen because the cell turnover rate of the oral mucosa epithelium is about ten to twelve days .
The comparison of these two cell collectors is of interest, since so far there have been very few investigations on different cell collectors in the literature. Previous studies compared, for example, a Cytobrush brush with a dermatological curette , with a wooden tongue spatula [39, 40] or with a metal spatula . In a study by Reboiras-López et al. three different sampling devices (Cytobrush, dermatological curette, OralCDx Brush) were used to perform liquid-based cytology . The used cell collectors (Orcellex® Brush, PapCone® Brush) differed in shape, the head of the PapCone® Brush was conical, and the head of the Orcellex® Brush was roll-shaped, resulting in a different approach to cell removal. Another significant difference was the texture of the cell collector heads, as the sponge was soft and pliable but the brush was rather firm and resistant. These characteristics influenced the cells collected during smear sampling and the appearance of the preparations. While taking the smears it was important to ensure a uniform, moderate contact pressure during rotation of the cell collector . In the case of disregard, this could lead, for example, to uneven removal of mucosal cells. In addition, smear sampling was always carried out by the same investigator, because there can be individual differences between various practitioners, although there are guidelines for correct smear sampling. Comparing the smears of the two used cell collectors with regard to their cytological appearance, the following observations can be summarized:
On the one hand, the PapCone® Brush was used in combination with conventional cell preparation. An advantage of this method was the rapid processing of the preparations in the laboratory of the section of Clinical and Experimental Oral Medicine after smear sampling by the direct rolling out of the sponges on the glass slides and the following air drying, fixation and staining. The glass slides produced by the conventional preparation can be examined immediately . Furthermore, the smear sampling with the softer texture of the spongy PapCone® Brush was more comfortable for the test person compared to the Orcellex® Brush. A disadvantage of the conventional exfoliative cytology was that irregular removal of the cell material was caused by the rolling-off process of the cell collector on the glass slides. By rolling-off the cell collector the cells were often concentrated in a line, while other areas were almost empty. In the areas with a lot of cell material, there were lots of cell overlapping, both overlaps of the cells by themselves and intersections with other cells. A further disadvantage of this conventional preparation technique was that, after rolling-off the collector, a considerable number of epithelial cells remained on the cell collectors. These remaining cells were discarded and thus were not included in the evaluation [19, 31]. Furthermore, some pollutant material, e.g. mucus, debris and blood cells, was present in the preparations. Such overlapping artifacts made the assessment and evaluation of the preparations partly difficult .
On the other hand, smear sampling was carried out using Orcellex® Brushes (Rovers Medical Devices B.V., Oss, Netherlands). The harder bristles of the cell smear brushes were noticeable for the test persons when taking the smear. In order to obtain sufficient cell material, twisting the brush ten times . After taking the smears, the brush heads were placed directly into the fixation liquid and washed out. For further processing, the Collection Vials (BD Diagnostics - TriPath, Erembodegem, Belgium) were sent to the laboratory of the Institute of Pathology (Mathias-Spital, Rheine), where the preparation of thin-layers was done by the procedure for SurePath™ preparations. A disadvantage of the liquid-based preparation method was the increased time required for packaging and delivery of the samples as well as the more elaborate processing for the liquid-based preparation in the laboratory. Thus, the liquid-based preparation of the smears was technically more complex, but showed significant benefits over conventional cytology. The advantages of the monolayer technique were that they did not have to be processed immediately. After placing the brush heads in the fixation liquid, storage of more than 24 h is possible. Another positive feature of this method is that the sampled cells are washed out of the brush by the liquid and are in the solution. Thus, it can be assumed that by washing out the cells from the cell collectors a significantly higher proportion of representative epithelial cells is present for the evaluation, since less important cell material is lost by liquid-based method than with conventional exfoliative cytology [7, 19, 31]. Failure sources due to incorrect transfer or fixation by the investigator can be minimized by the monolayer preparation technique . When the liquid was applied to the glass slide, no mechanical deformation took place and the cells retained their original shape. Furthermore, this procedure also resulted in a much more even distribution of the cells on the slide and thus less cell overlapping was found [2, 7, 19, 20, 31]. Another important advantage of this method was that the unwanted harvested material, such as blood cells, debris and mucus, was almost completely filtered out before being transferred to the slides [2, 19, 20, 31]. Therefore, the preparations showed less impurities and the background appeared clearer [7, 19]. Thus, an evaluation of these preparations proved to be much simpler and more accurate, since the cells were homogeneously distributed and almost without overlaps. In conclusion, both the conventional exfoliative cytology and the liquid-based method provide reliable results. However, the liquid-based preparation method showed obvious advantages, because it represented the cell morphology better and, due to the aforementioned advantages, it was superior to the conventional cytology [7, 38, 43]. Thus the liquid-based cytology not only provided better results, but also cell material was available for further investigation [38, 43].
In summary, qualitative and quantitative differences in the preparations were detected, depending on the cell collector used and the cell preparation method. There was a superiority of the Orcellex® Brush in combination with the liquid-based monolayer technique.
We acknowledge support from the German Research Foundation (DFG) and Leipzig University withinthe program of Open Access Publishing.
Availability of data and materials
The datasets used and/or analysed during the current study are available from the corresponding author on reasonable request.
CO: contributed to data acquisition, design, analysis and interpretation, writing of manuscript, NH: contributed to data acquisition, HN: contributed to data acquisition, MYD: statistical analysis, TWR: contributed design, critically revised the manuscript, All authors gave final approval and agree to be accountable for all aspects of the work.
Ethics approval and consent to participate
All procedures performed in studies involving human participants were in accordance with the ethical standards of the institutional and/or national research committee and with the 1964 Helsinki declaration and its later amendments or comparable ethical standards. This study was endorsed by the Leipzig Ethics Committee (file: 030–14-27012014).
Consent for publication
Informed consent was obtained from all individual participants included in the study.
The authors declare that they have no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
- Remmerbach TW, Hemprich A, Bocking A. Minimalinvasive Bürstenbiopsie: innovative Methode zur Früherkennung oraler Plattenepithelkarzinome. Schweiz Monatsschr Zahnmed. 2007;117:926–40.PubMedGoogle Scholar
- Kujan O, Desai M, Sargent A, Bailey A, Turner A, Sloan P. Potential applications of oral brush cytology with liquid-based technology: results from a cohort of normal oral mucosa. Oral Oncol. 2006;42:810–8.View ArticlePubMedGoogle Scholar
- Remmerbach TW, Weidenbach H, Pomjanski N, Knops K, Mathes S, Hemprich A, Bocking A. Cytologic and DNA-cytometric early diagnosis of oral cancer. Anal Cell Pathol. 2001;22:211–21.View ArticlePubMedPubMed CentralGoogle Scholar
- Macey R, Walsh T, Brocklehurst P, Kerr AR, Liu JLY, Lingen MW, et al. Diagnostic tests for oral cancer and potentially malignant disorders in patients presenting with clinically evident lesions. Cochrane Database Syst Rev. 2015;29(5):CD010276. https://doi.org/10.1002/14651858.CD010276.pub2.
- Pandey P, Agarwal S, Ralli M, Dixit A, Singh D. Oral brush liquid-based cytology: a study of concordance between a cytotechnologist and a Cytopathologist. Acta Cytol. 2018;62:121–29.Google Scholar
- H Alsarraf A, Kujan O, Farah CS. The utility of oral brush cytology in the early detection of oral cancer and oral potentially malignant disorders: a systematic review. J Oral Pathol Med. 2018;47:104–16. https://doi.org/10.1111/jop.12660.View ArticlePubMedGoogle Scholar
- Hayama FH, Motta ACF, Silva Ade P, Migliari DA. Liquid-based preparations versus conventional cytology: specimen adequacy and diagnostic agreement in oral lesions. Med Oral Patol Oral Cir Bucal. 2005;10:115–22.PubMedGoogle Scholar
- Kobayashi TK, Ueda M, Nishino T, Terasaki S, Kameyama T. Brush cytology of herpes simplex virus infection in oral mucosa: use of the ThinPrep processor. Diagn Cytopathol. 1998;18:71–5.View ArticlePubMedGoogle Scholar
- Maksem JA, Weidmann J. Specialized preparative devices are not needed for liquid-based, thin-layer cytology: an alternate manual method using a metastable alcoholic gel. Diagn Cytopathol. 2001;25:262–4.Google Scholar
- Mollaoglu N, Wilson MJ, Cowpe JG. Extraction of DNA from oral cytological samples by scraping and smear method suitable for restriction site mutation analysis: a pilot study. Diagn Cytopathol. 2001;25:83–5.View ArticlePubMedGoogle Scholar
- Driemel O, Kunkel M, Hullmann M, Kleinsasser N, Staudenmaier R, Muller-Richter U, et al. Performance of conventional oral brush biopsies. HNO. 2008;56:205–10. https://doi.org/10.1007/s00106-007-1662-1. View ArticlePubMedGoogle Scholar
- Driemel O, Dahse R, Hakim SG, Tsioutsias T, Pistner H, Reichert TE, Kosmehl H. Laminin-5 immunocytochemistry: a new tool for identifying dysplastic cells in oral brush biopsies. Cytopathology. 2007;18:348–55. https://doi.org/10.1111/j.1365-2303.2006.00401.x.View ArticlePubMedGoogle Scholar
- Smith JHF. Cytology, liquid-based cytology and automation. Best Pract Res Clin Obstet Gynaecol. 2011;25:585–96. https://doi.org/10.1016/j.bpobgyn.2011.04.006.View ArticlePubMedGoogle Scholar
- Bishop JW, Bigner SH, Colgan TJ, Husain M, Howell LP, McIntosh KM, et al. Multicenter masked evaluation of AutoCyte PREP thin layers with matched conventional smears. Including initial biopsy results Acta Cytol. 1998;42:189–97.Google Scholar
- McGoogan E, Reith A. Would monolayers provide more representative samples and improved preparations for cervical screening? Overview and evaluation of systems available. Acta Cytol. 1996;40:107–19.View ArticlePubMedGoogle Scholar
- Vassilakos P, Cossali D, Albe X, Alonso L, Hohener R, Puget E. Efficacy of monolayer preparations for cervical cytology: emphasis on suboptimal specimens. Acta Cytol. 1996;40:496–500.View ArticlePubMedGoogle Scholar
- Howell LP, Davis RL, Belk TI, Agdigos R, Lowe J. The AutoCyte preparation system for gynecologic cytology. Acta Cytol. 1998;42:171–77.Google Scholar
- Maurer K. Oral brush biopsy analysis by MALDI-ToF mass spectrometry for early cancer diagnosis [dissertation]: Medizinische Fakultät, Universität Leipzig. 2013.Google Scholar
- Buch J. Evaluation der flüssigkeitsbasierten (Dünnschicht-) Zytologie oraler Bürstenbiopsien zur Dignitätsabklärung von unklaren Mundschleimhautveränderungen [dissertation]: Medizinische Fakultät, Universität Leipzig. 2016.Google Scholar
- Goodson ML, Sloan P, Wadehra V, Johnson SJ, Robinson CM, Aubourg G, Thomson PJ. Orcellex® brush biopsy and liquid-based cytology - assessment of a new diagnostic technique in oral potentially malignant disorder management. J Oral Maxillofac Surg Photon. 2014;117:224–31.Google Scholar
- Reboiras-López MD, Pérez-Sayans M, Somoza-Martín JM, Antúnez-López JR, Gándara-Vila P, Gayoso-Diz P, et al. Comparison of three sampling instruments, Cytobrush, curette and OralCDx, for liquid-based cytology of the oral mucosa. Biotech Histochem. 2012;87:51–8.View ArticlePubMedGoogle Scholar
- Goodson ML, Smith DR, Thomson PJ. Efficacy of oral brush biopsy in potentially malignant disorder management. J Oral Pathol Med. 2017;46:896–901. https://doi.org/10.1111/jop.12627.PubMedView ArticleGoogle Scholar
- Kujan O, Pemberton MN, Schwarz M, Sloan P. Evaluation of an innovative oral brush for potential applications using liquid based cytology. J Oral Sci. 2018;60:45-50. https://doi.org/10.2334/josnusd.16-0702.
- Hegde V, Nambiar S, Yadav N. Efficacy of centrifuged liquid-based cytology over conventional cytology: a comparative study. J Cancer Res Ther. 2017;13:951–55. https://doi.org/10.4103/0973-1482.177219.
- Remmerbach TW, Weidenbach H, Muller C, Hemprich A, Pomjanski N, Buckstegge B, Bocking A. Diagnostic value of nucleolar organizer regions (AgNORs) in brush biopsies of suspicious lesions of the oral cavity. Anal Cell Pathol. 2003;25:139–46.View ArticlePubMedPubMed CentralGoogle Scholar
- Koss LG. The Papanicolaou test for cervical cancer detection. A triumph and a tragedy. JAMA. 1989;261:737–43.View ArticlePubMedGoogle Scholar
- Pittler MH, Blümle A, Meerpohl JJ, Antes G. CONSORT 2010 statement: updated guidelines for reporting parallel group randomised trials. Dtsch Med Wochenschr. 2011;136:20–3.View ArticleGoogle Scholar
- World Medical Association WM. WMA Deklaration von Helsinki - Ethische Grundsätze für die medizinische Forschung am Menschen: Verabschiedet von der 18. WMA-Generalversammlung. In: Juni 1964 Helsinki (Finnland); Oktober; 2013.Google Scholar
- Bloching M, Reich W, Schubert J, Grummt T, Sandner A. Micronucleus rate of buccal mucosal epithelial cells in relation to oral hygiene and dental factors. Oral Oncol. 2008;44:220–26.Google Scholar
- Reich W. Der Ames- und Mikrokerntest als Möglichkeiten zur Früherkennung kanzerogener Exposition in der Mundhöhle und ihre Beziehung zum dentalen Status [Dissertation]. Martin-Luther-Universität Halle-Wittenberg. 2005.Google Scholar
- Reichardt T, Remmerbach TW. Tumoren in der Mundhöhle – zahnärztliche Aspekte der Früherkennung. Zahnmedizin up2date. 2016;10:67–83.Google Scholar
- Klöppel G, Kreipe HH, Remmele W, editors. Pathologie: Zytopathologie. 3rd ed. Berlin. Heidelberg: Springer-Verlag, S; 2011. p. 605–41.View ArticleGoogle Scholar
- Thomas C, editor. Histopathologie. Lehrbuch und Atlas zur allgemeinen und speziellen Pathologie. 13. Stuttgart: Schattauer GmbH; 2001. p. 411–36.Google Scholar
- Remmele W, editor. Pathologie: Band 8: Zytopathologie. New York: Springer-Verlag. 2000; p. 425–55.Google Scholar
- TriPath Imaging. Produktbeilage. SurePath collection product insert. USA: Burlington, NC 27215; 2009.Google Scholar
- Bocking A. Standardization of cytopathologic diagnosis. Pathologe. 1998;19:236–41.View ArticlePubMedGoogle Scholar
- Hellwig E, Klimek J, Attin T. Einführung in die Zahnerhaltung: Prüfungswissen Kariologie, Endodontologie und Parodontologie. In: Freiburg, Gießen. 6th ed. Zürich: Deutscher Zahnärzte Verlag. 2013; p. 457–58.Google Scholar
- Navone R. Cytology of the oral cavity: a re-evaluation. Pathologica. 2009;101:6–8.PubMedGoogle Scholar
- Jones AC, Pink FE, Sandow PL, Stewart CM, Migliorati CA, Baughman RA. The Cytobrush plus cell collector in oral cytology. Oral Surg Oral Med Oral Pathol. 1994;77:95–9.View ArticlePubMedGoogle Scholar
- Ogden GR, Cowpe JG, Green M. Cytobrush and wooden spatula for oral exfoliative cytology. A comparison. Acta Cytol. 1992;36:706–10.Google Scholar
- Queiroz JB, Lima CF, Burim RA, Brandao AA, Cabral LA, Almeida JD. Exfoliative cytology of the oral mucosa: comparison of two collection methods. Gen Dent. 2010;58:196–9.Google Scholar
- Remmerbach TW, Maurer K, Janke S, Schellenberger W, Eschrich K, Bertolini J, et al. Oral brush biopsy analysis by matrix assisted laser desorption/ionisation-time of flight mass spectrometry profiling--a pilot study. Oral Oncol. 2011;47:278–81. https://doi.org/10.1016/j.oraloncology.2011.02.005.View ArticlePubMedGoogle Scholar
- Navone R, Burlo P, Pich A, Pentenero M, Broccoletti R, Marsico A, Gandolfo S. The impact of liquid-based oral cytology on the diagnosis of oral squamous dysplasia and carcinoma. Cytopathology. 2007;18:356–60.View ArticlePubMedGoogle Scholar