- Short report
- Open Access
- Open Peer Review
This article has Open Peer Review reports available.
Strain driven fast osseointegration of implants
© Joos et al; licensee BioMed Central Ltd. 2005
Received: 03 May 2005
Accepted: 01 September 2005
Published: 01 September 2005
Although the bone's capability of dental implant osseointegration has clinically been utilised as early as in the Gallo-Roman population, the specific mechanisms for the emergence and maintenance of peri-implant bone under functional load have not been identified. Here we show that under immediate loading of specially designed dental implants with masticatory loads, osseointegration is rapidly achieved.
We examined the bone reaction around non- and immediately loaded dental implants inserted in the mandible of mature minipigs during the presently assumed time for osseointegration. We used threaded conical titanium implants containing a titanium2+ oxide surface, allowing direct bone contact after insertion. The external geometry was designed according to finite element analysis: the calculation showed that physiological amplitudes of strain (500–3,000 ustrain) generated through mastication were homogenously distributed in peri-implant bone. The strain-energy density (SED) rate under assessment of a 1 Hz loading cycle was 150 Jm-3 s-1, peak dislocations were lower then nm.
Bone was in direct contact to the implant surface (bone/implant contact rate 90%) from day one of implant insertion, as quantified by undecalcified histological sections. This effect was substantiated by ultrastructural analysis of intimate osteoblast attachment and mature collagen mineralisation at the titanium surface. We detected no loss in the intimate bone/implant bond during the experimental period of either control or experimental animals, indicating that immediate load had no adverse effect on bone structure in peri-implant bone.
In terms of clinical relevance, the load related bone reaction at the implant interface may in combination with substrate effects be responsible for an immediate osseointegration state.
Although the bone's capability of dental implant osseointegration has clinically been utilised as early as in the Gallo-Roman population , the specific mechanisms for the emergence and maintenance of peri-implant bone under functional load have not been identified. Here we show that under immediate loading of specially designed dental implants with masticatory loads, osseointegration is rapidly achieved. As the osseointegration state is much faster reached than commonly assumed, osseointegration is a strain dependant highly dynamic process.
Osseointegration is defined as a direct and stable anchorage of an implant by the formation of bony tissue without growth of fibrous tissue at the bone-implant interface . A defining feature of osseointegration is that osteoblasts and mineralized matrix contacts the implant surface even when loads are applied. A common perception is that several weeks must be given to achieve implant osseointegration.
Bone response on an implant surface depends on the reaction of cells and matrix towards the material surface as well as to the mechanical constraints in the vincinity of the implant. The maintenance of bone and its adaptation to external loads is based on a complex strain driven regulatory process of cells and matrix components [5, 6]. Outside-in mechanical tension exert direct effects on cell behaviour by activating biochemical signalling pathways and regulating gene expression through focal adhesions . Frost  provided a paradigm for the mechanical control of cellular bone modelling, the process whereby bone is laid down onto surfaces without necessarily preceded by resorption. Recent investigations have indicated that the strain related bone modelling process is also regulative for bone tissue formation in healing tissue .
Using an atomic force microscope, a molecular mechanistic origin for the remarkably fast recovery of toughness after bone deformation was found, when deformation of less then 50 nm at the surface of multivalent ions (as in the case of Ti-oxide) is present . Our understanding of osseointegration theorises that bone strengthening responds to a highly specific mechanical design. Even if long-term osseointegrated implants show what seems to be similar bone tissue reactions, osseointegration might be able to be achieved more rapidly than otherwise observed. Screw type titanium implants, as used in dental implantology, have in contrast to orthopaedic implants not only been convincingly shown very good clinical long-term success , but were also successful when load transfer is immediately present as seen in traumatology. In terms of clinical relevance, the load related bone reaction at the implant interface may in combination with substrate effects be responsible for an immediate osseointegration state.
- Crubezy E, Murail P, Girard L, Bernadou JP: False teeth of the Roman world. Nature. 1998, 391: 29. 10.1038/34067.View ArticlePubMedGoogle Scholar
- Branemark PI, Adell R, Breine U, Hansson BO, Lindstrom J, Ohlsson A: Intra-osseous anchorage of dental prostheses. I. Experimental studies. Scand J Plast Reconstr Surg. 1969, 3: 81-100.View ArticlePubMedGoogle Scholar
- Okumura A, Goto M, Goto T, Yoshinari M, Masuko S, Katsuki T, Tanaka T: Substrate affects the initial attachment and subsequent behavior of human osteoblastic cells (Saos-2). Biomaterials. 2001, 22: 2263-2271. 10.1016/S0142-9612(00)00415-4.View ArticlePubMedGoogle Scholar
- Turner CH, Takano Y, Owan I: Aging changes mechanical loading thresholds for bone formation in rats. J Bone Miner Res. 1995, 10: 1544-1549.View ArticlePubMedGoogle Scholar
- Rubin CT, Pratt GW, Porter AL, Lanyon LE, Poss R: Ultrasonic measurement of immobilization-induced osteopenia: an experimental study in sheep. Calcif Tissue Int. 1988, 42: 309-312.View ArticlePubMedGoogle Scholar
- Huiskes R, Ruimerman R, van Lenthe GH, Janssen JD: Effects of mechanical forces on maintenance and adaptation of form in trabecular bone. Nature. 2000, 405: 704-706. 10.1038/35015116.View ArticlePubMedGoogle Scholar
- Chicurel ME, Singer RH, Meyer CJ, Ingber DE: Integrin binding and mechanical tension induce movement of mRNA and ribosomes to focal adhesions. Nature. 1998, 392: 730-733. 10.1038/33719.View ArticlePubMedGoogle Scholar
- Frost HM: The mechanostat: a proposed pathogenic mechanism of osteoporoses and the bone mass effects of mechanical and nonmechanical agents. Bone Miner. 1987, 2: 73-85.PubMedGoogle Scholar
- Frost HM: A brief review for orthopaedic surgeons: Fatigue damage (microdamage in bone (its determinants and clinical implications). J Orthop Sci. 1998, 3: 272-281. 10.1007/s007760050053.View ArticlePubMedGoogle Scholar
- Thompson JB, Kindt JH, Drake B, Hansma HG, Morse DE, Hansma PK: Bone indentation recovery time correlates with bond reforming time. Nature. 2001, 13: 773-776. 10.1038/414773a.View ArticleGoogle Scholar
- Albrektsson T, Johansson : Osteoinduction, osteoconduction and osseointegration. C Eur Spine J. 2001, 10: 96-101. 10.1007/s005860100282.View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.