Distraction osteogenesis in a severe mandibular deficiency
© Ortakoglu et al; licensee BioMed Central Ltd. 2007
Received: 28 June 2006
Accepted: 20 January 2007
Published: 20 January 2007
Distraction osteogenesis is an alternative treatment method for the correction of mandibular hypoplasia. In this case report, distraction with a multidirectional extraoral device was performed to gradually lengthen the corpus and ramus of a patient who had a severe hypoplastic mandible.
Materials and methods
The patient underwent bilateral extraoral ramus and corpus distraction osteogenesis. After seven days of latency period, distraction was performed 0.5 mm twice a day. Subsequent consolidation period was 12 weeks.
The patient's mandible was elongated successfully. Cephalometric analysis revealed that ANB angle decreased from 13° to 6°, overjet of 15 mm decreased to 4 mm, corpus length increased from 49 mm to 67 mm, and ramus length increased from 41 mm to 43 mm. Posterior airway space (PAS) also increased due to advancement of the mandible. In stereolithographic model evaluation it was determined that the distances from condylion to gonion and from gonion to pogonion increased.
Satisfactory results from both aesthetic and functional standpoints were obtained by distraction osteogenesis of the ramus and corpus.
Distraction osteogenesis (DO) is a biologic process of new bone formation between the surfaces of bone segments that are gradually separated by incremental traction . A callus forms between the separated bone segments and as long as the traction proceeds, callus tissues are stretched inducing the new bone formation .
DO was first introduced by Codivilla at the beginning of twentieth century and during 1950s, the studies of Ilizarov made a contribution in the development of the technique by elucidating the biological and mechanical principals in the formation of new bone [3–5]. DO has been applied to craniofacial region since McCarthy et al  reported the first clinical application of the technique in the treatment of four children with either unilateral or bilateral mandibular hypoplasia.
In the lengthening of the hypoplastic mandible, external or intraoral devices have been used. Although intraoral distractors are more convenient socially and leave no residual skin scars, extraoral bidirectional or multidirectional devices should be preferred in severe hypoplastic cases, when three-dimensional vector control is required and gonial angle control is necessary .
In this case report, we intended to present the treatment of an adult patient who had severe mandibular deficiency. An extraoral multidirectional distractor was used to achieve independent horizontal distraction of the corpus and vertical distraction of the ramus. Amount of lengthening was determined with cephalograms and stereolithographic models. Stereolithographic modelling is a well known system Moreover alterations at the posterior airway space due to elongation of mandible were also presented.
A 23-year old male patient who had severe mandibular hypoplasia referred to the Department of Oral and Maxillofacial Surgery for treatment. His main complaints were unaesthetic appearance, snoring, wheezing, and difficulties during respiration, speech, and chewing.
To determine the skeletal deformity, cephalometric analysis and measurements on stereolithographic three dimensional skull prototype model obtained through three dimensional computerized tomographic images of the patient were used. The stereolithographic model was built by Spectrum Z510 Color 3D Printers (Z Corporation-Burlington, USA).
Cephalometric analysis of the patient
-3 mm/-2 mm
-4 mm/-3 mm
Ramus and corpus lengths were also measured on the stereolithographic models. The preoperative distance between condyle and gonion was 50 mm and the distance between gonion and menton was 69 mm for the right side. They were 44 mm and 68 mm, respectively for the left side. The patient had severe mandibular hypoplasia due to inadequate ramus and corpus length, therefore using an extraoral multiplanar distractor was inevitable. Tritrac® External Distractor (Ucmed Medical Ltd. Co.-Ankara, Turkey) was used for the treatment of the presented case.
Orthodontic treatment, surgical technique and distraction procedure
Preadjusted appliances (0.018 × 0.022 inch) were placed in the maxillary and mandibular arch and open coil spring was used to open space for the left lateral incisor. Following the leveling phase, 0.016 × 0.022 inch arch wires were placed on the upper and lower teeth.
Because of the limitation of the mouth opening, nasotracheal intubation was performed with fiberoptic bronchoscope. Under general anaesthesia, after completing the intraoral incisions and exposing the cortex of mandibular corpus and ramus on both sides, the device was secured to the mandible via 6 percutaneous pins. This system had 3 pairs of pins to be secured; one pair to the ramus, the other to the angulus and the last one to the corpus region inferior to the mandibular canal. During this procedure extraoral incisions and trochar were used. The mandibular osteotomy was then performed with round and fissure burs and osteotomes. Two osteotomies on each ipsilateral side were planned. One of the osteotomies was on the ramus and the other one was anterior to the second molar. After the completion of the osteotomies, the distractors were tested and were placed paralell to each other. Then, the incisions were primarily closed.
A 7 days of latency phase was waited for callus formation. Afterwards, distraction was performed at a rate of 0.5 mm, twice a day. Ramus was distracted for 7 days and corpus was distracted for 45 days. While the corpus was being distracted, the angles of distractors were reduced 5° on days 10, 20, 30, 40 so as to decrease the gonial angle. Paralellization of the distractors were protected during this procedure. Once the desired bone length had been acquired, 3 months of consolidation was allowed. Following the consolidation phase the distractor was removed under local anaesthesia and the extraoral wounds were debrided and primarily closed and left for recovery.
The patient used class II intraoral elastics at night time in order to prevent relapse and elastic traction was completed after 3 months. The patient was followed-up for one more month in order to determine whether relapse would occur. Brackets were debonded 4 months after consolidation period. The total treatment time was 14 months.
Finally the prosthetic rehabilitations of the missing teeth in the mandible were solved with stationary restorations.
Cephalometric analysis revealed that overjet of 15 mm and overbite of 4 mm decreased to 4 mm and 1 mm, respectively. ANB angle decreased from 13° to 6°, distance between pogonion and NV increased from 42 mm to 24 mm, corpus length (Go-Gn) increased from 49 mm to 67 mm, and ramus length (Ar-Go) increased from 41 mm to 43 mm showing the successful elongation of ramus and corpus. Gonial angle, inclination of the mandibular plane, and Y-axis decreased revealing the anterior rotation of the mandible. Elongation of the ramus length increased the posterior face height and anterior rotation of the mandible decreased the anterior face height. These alterations changed the profile of the patient and provided more symmetric and aesthetic appearance. Cephalometric superimpositions of total face on SN plane is demonstrated in Figure 4, and and pre-treatment and post-treatment cephalometric analysis are presented in Table 1.
PAS analysis of the patient
Mandibular hypoplasia is one of the most common malformations of the facial skeleton. It is usually associated with a deficient gonial angle, ascending ramus, and mandibular corpus. Maxilla, zygoma, temporal bone, cranial vault and cervical spine are the other anatomic landmarks that may be affected [1, 7]. Pruzansky  classified mandibular hypoplasia according to the severity of the deformity. In this classification, as the severity increases the ascending ramus is affected to a greater degree.
In the presented case, the patient had a severe mandibular hypoplasia due to short ramus and mandibular body. Since mandibular lengthening procedure required vertical and horizontal components of distraction, an extraoral multiplanar distractor was selected. Intraoral devices usually apply unidirectional force due to the limited space in the oral cavity. Problems such as malocclusion and deficiencies in lower facial contours have been encountered in use of unidirectional devices for mandibular distraction . Extraoral bidirectional or multidirectional distractors should be preferred in the correction of severe deformities including ramus, corpus and the angle of mandible [9, 10].
Molina and Ortiz-Monasterio  were the first to use bidirectional osteodistraction in the mandible to lengthened both ramus and corpus of the mandible simultaneously. Anatomically, the mandible consists of two halves that are fused at an acute angle in the midline forming V-shaped bone structure. Therefore, in order to correct severe mandibular deformities in three dimensions, independent lengthening of mandibular corpus and ramus must be combined with gradual angular adjustments. Because of this, several multidirectional extraoral distractors have been developed in the last decade [11, 12]. Multidirectional distractors are advantageous in distraction of the mandible in all three planes of space . Linear or angular correction in the sagittal and vertical planes and angular correction in the transverse plane can be obtained with a multidirectional distractor and movement of bone segments or shape of the regenerate bone may be changed during distraction process . In our patient, we used a multidirectional distractor and performed two ipsilateral osteotomies to shorten the total distraction period by creating two callus sites. This also allowed the development of a mandibular angle [1, 2, 12]. In the cephalometric analysis and model measurements we couldn't observe a significant increase in ramus length. Although the vertical component of the distractor was activated for 7 days, it did not reflect to ramus length exactly. Similarly, horizontal component was activated 45 mm but corpus length increased only 18 mm according to the cephalometric analysis. In our opinion the reason of the relapse, which was more than expected, was probably depending on the angular alterations performed during distraction period on days 10, 20, 30 and 40. These angular alterations were made in order to decrease the gonial angle and create anterior rotation of the mandible.
In the treatment of the patient, distraction osteogenesis was preferred as orthognathic surgery has relapse risk in severe mandibular deficiencies requiring lengthening of the mandible more than 8–10 mm. Even when the surgical technique is modified, 15 mm is approximately the outer limit of predictable surgical mandibular advancement. In our patient, lengthening of the ramus was also needed but after conventional orthognathic surgery procedures, pterygoid muscle usually does not adapt to the elongation of ramus. However, during distraction osteogenesis, active histiogenesis occurs in different tissues including gingiva, blood vessels, ligaments, cartilage, muscles and nerves [15, 16]. These adaptive changes in the soft tissues decrease the relapse risk and allow the treatment of severe facial deformities.
DO has also some risks such as infection, loosening of the distractor, paraesthesia, and excessive skin damage caused by the pins of the extraoral device. Strategic mistakes such as inappropriate distractor configuration or inadequate calculation of distraction parameters and technical mistakes like misalignment of the distractor leading to displacement of the bone segment, insufficient rate of the lengthening, premature consolidation may cause undesired results. These complications are usually related with the experience of the surgeon [17–19]. In the presented case, only facial scar was observed that was the inevitable result of the extraoral distractor.
Severe mandibular hypoplasia can lead to a reduction of oropharyngeal capacity and glossoptosis because of the posterior location of the insertion of the suprahyoid muscles into the mandible. As a result, upper airway obstruction, feeding difficulties, gastroesophageal reflux may occur. Several authors have reported that these conditions could be resolved by the help of mandibular distraction [20–22]. Similarly, following the advancement of the mandible, respiratory problems, snoring, and difficulties during feeding and talking improved in our patient, due to the increase in the PAS and correction of the excessive overjet.
Elongation of the corpus length increased the volume of the oral cavity therefore tongue was relieved. Before treatment, tongue tip was positioned over the incisal edges of the lower incisors because of insufficient space. However, following the treatment, it retruded behind these teeth to a normal position. This retrusion decreased the value of TL (distance between tongue tip and Eb). The tongue accompanied the advancement of the mandible and postural alterations occurred. Since the tongue moved in the anterior direction, all the PAS parameters increased. The most significant increase was in the N-PAS distance which was threefold higher than the initial value. Translation of the tongue affected not only the width of the N-PAS but also changed the position of this area. N-PAS was under I-PAS before treatment but it positioned between S-PAS and I-PAS following the treatment and this was the place where it should be. During the distraction of the corpus, mandible revealed anterior rotation and probably this rotation affected the hyoid bone through the muscles and caused an elevation at the hyoid so the perpendicular distance from hyoid to mandibular plane decreased after treatment.
During the treatment planning, both cephalometric analysis and solid model measurements were utilized. Solid model revealed the exact shape and size of the skeletal structures and provided the advantage of measuring the length of ramus and corpus directly and separately for right and left sides. Model surgery may be performed on the solid model so as to determine the osteotomy sides, inclination of the osteotomy and distraction vector. Distractors may be placed on the model after the model is cut at the osteotomy side with a saw and it may be activated to observe the results that would occur at the end of treatment and to determine the distraction period. In our opinion solid models could be used successfully in the treatment planning of distraction osteogenesis or orthognathic surgery cases.
We are very grateful to 4C Engineering, Istanbul-Turkey, for their helps in providing us the 3D solid models.
- Samchukov ML, Cope JB, Cherkashin AM: Craniofacial distraction osteogenesis. 2001, St.Louis: MosbyGoogle Scholar
- Klein C, Howaldt HP: Correction of mandibular hypoplasia by means of bi-directional callus distraction. J Craniofac Surg. 1996, 7: 258-266. 10.1097/00001665-199607000-00002.View ArticlePubMedGoogle Scholar
- Codivilla A: On the means of lengthening, in the lower limbs, the muscles and tissues which are shortened through deformity. Clin Orthop Relat Res. 1994, (301): 4-9.
- Ilizarov GA: The tension-stress effect on the genesis and growth of tissues: Part I. The influence of stability of fixation and soft-tissue preservation. Clin Orthop Relat Res. 1989, 238: 249-281.PubMedGoogle Scholar
- Ilizarov GA: The tension-stress effect on the genesis and growth of tissues: Part II. The influence of the rate and frequency of distraction. Clin Orthop Relat Res. 1989, 239: 263-285.PubMedGoogle Scholar
- McCarthy JG, Schreiber J, Karp N, Thorne CH, Grayson BH: Lengthening of the human mandible by gradual distraction. Plast Reconst Surg. 1992, 89: 1-10.View ArticlePubMedGoogle Scholar
- Pruzansky S: Not all dwarfed mandibles are alike. Birth defects. 1969, 1: 120-Google Scholar
- Pensler JM, Goldberg DP, Lindell B, Carroll NC: Skeletal distraction of the hypoplastic mandible. Ann Plast Surg. 1995, 34: 130-137. 10.1097/00000637-199502000-00004.View ArticlePubMedGoogle Scholar
- Molina F, Ortiz-Monasterio F: Mandibular elongation and remodelling by distraction: a farewell to major osteotomies. Plast Reconst Surg. 1995, 96: 825-842.View ArticlePubMedGoogle Scholar
- Hollier LH, Rowe NM, Mackool RJ, Williams JK, Kim JH, Longaker MT, Grayson BH, McCarthy JG: Controlled multiplanar distraction of the mandible. Part III: Laboratory studies of sagittal (anteroposterior) and horizontal (mediolateral) movements. J Craniofac Surg. 2000, 11: 83-95.View ArticlePubMedGoogle Scholar
- McCarthy JG: Craniofacial microsomia. A primary or secondary surgical treatment program. Clin Plast Surg. 1997, 24: 459-474.PubMedGoogle Scholar
- McCarthy JG, Williams JK, Grayson BH, Crombie JS: Controlled multiplanar distraction of the mandible: device development and clinical application. J Craniofac Surg. 1998, 9: 322-329.View ArticlePubMedGoogle Scholar
- Gateno J, Teichgraeber JF, Aguilar E: Distraction osteogenesis: A new surgical technique for use with the multiplanar mandibular distractor. Plast Reconst Surg. 2000, 105: 883-888. 10.1097/00006534-200003000-00009.View ArticlePubMedGoogle Scholar
- Maull DJ: Review of devices for distraction osteogenesis of the craniofacial complex. Semin Orthod. 1999, 5: 64-73. 10.1016/S1073-8746(99)80045-0.View ArticlePubMedGoogle Scholar
- Hawkinson RT: Retrognathia correction by means of an arcing osteotomy in the ascending ramus. J Prosthet Dent. 1968, 20: 77-86. 10.1016/0022-3913(68)90214-X.View ArticlePubMedGoogle Scholar
- Liou EJ, Huang CS: Rapid canine retraction through distraction of the periodontal ligament. Am J Orthod Dentofacial Orthop. 1998, 114: 372-382. 10.1016/S0889-5406(98)70181-7.View ArticlePubMedGoogle Scholar
- Dahl MT, Gulli B, Berg T: Complications of limb lengthening. A learning curve. Clin Orthop Relat Res. 1994, 301: 10-18.PubMedGoogle Scholar
- Hughes GB: The learning curve in stapes surgery. Laryngoscope. 1991, 101: 1280-1284.View ArticlePubMedGoogle Scholar
- Suhr MA, Kreusch T: Technical considerations in distraction osteogenesis. Int J Oral Maxillofac Surg. 2004, 33: 89-94. 10.1054/ijom.2003.0462.View ArticlePubMedGoogle Scholar
- Mandell DL, Yellon RF, Bradley JP, Izadi K, Gordon CB: Mandibular distraction for micrognathia and severe upper airway obstruction. Arch Otolaryngol Head Neck Surg. 2004, 130: 344-348. 10.1001/archotol.130.3.344.View ArticlePubMedGoogle Scholar
- Monasterio FO, Molina F, Berlanga F, Lopez ME, Ahumada H, Takenaga RH, Ysunza A: Swallowing disorders in Pierre Robin Sequence: It's correction by distraction. J Craniofac Surg. 2004, 15: 934-941. 10.1097/00001665-200411000-00009.View ArticlePubMedGoogle Scholar
- Wang X, Wang XX, Liang C, Yi B, Lin Y, Li ZL: Distraction osteogenesis in correction accompanying obstructive sleep apnea syndrome. Plast Reconst Surg. 2003, 112: 1549-1559. 10.1097/01.PRS.0000086734.74795.C4.View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.